# Current sensor with Arduino

The following project was designed for a need: with my friend Davide I’m working on a model railway (Il Plastico di Davide e Luca): we needed to know how much current it uses to size our power supplies (booster)…

### The sensor

For current measurement, I used ACS712 sensor from Allegro MicroSystems (here’s its datasheet). You can find 3 models of this integrated circuit, depending on the maximum measured current (5-20-30A); for our requirements I chose the 5A model.

ACS712 is very easy to use: voltage across VIOUT and ground PIN is proportional to current flowing between IP+ e IP-. It requires a power supply of 5V (VCC) and two capacitors to filter power supply and output; on its datasheet you can find the typical application:

The chip is sold in a SOIC package; SparkFun sells a convenient breakout board with ACS712 presoldered. I connected that board to a perfboard where I soldered the two capacitors and three PINs for Arduino connection:

### Measurement

As I wrote, output voltage is proportional to input current. The value that links the two measurements is sensitivity (you can find it on the datasheet) which – for 5A model – has a typical value of 185mV/A. The sensor can measure positive and negative currents (range -5A…5A), so if input current is 0, output value is 2.5V.

We’re going to read output value with one of the analog inputs of Arduino and its analogRead() function. That function outputs a value between 0 (0V in input) and 1023 (5V in input) that is 0,0049V for each increment.

The formula you need to convert units from analogRead() to Ampere is:

### Arduino

Arduino sketch is very simple: it reads the value on AN0 PIN, converts it with the formula above and sends the result to a Personal Computer via serial connection. My PC is running AnalogDemo software which plots data to a chart and updates a  feed on Pachube.

ACS712 measures an instant value; my first tests showed that the measure was too floating, that’s the reason why the sketch below reads 1000 values, computes an average and sends it to my PC. analogRead() needs about 0.1ms for each reading, if you want a value for each second, you need a delay of 1ms in the loop:

```void setup() {   Serial.begin(9600); }   void loop() {   float average = 0; for(int i = 0; i < 1000; i++) { average = average + (.0264 * analogRead(A0) -13.51) / 1000; delay(1); } Serial.println(average); }```

Test

To test the accuracy of my sensor, I used a microcontrolled battery charger, with the ACS712 between its outputs and the battery pack in charging:

I was able to found a very close match between the output current shown by the charger and the values read by the current sensor.

### Get on the train!

Last Saturday we used the circuit with our model railway, here you can read about the test and watch a short video with the circuit running during a small short-circuit.

### Notes

If you read the sensor’s output with Arduino, you have a resolution of about 0,026A (the value you get from  analogRead() increases by a unit every 0.026A). To improve the resolution, you can add an opAmp which extends the output range; SparkFun again sells a breakout board with that improvement.

#### Related Posts

Tuesday November 29th, 2011 - 11:51 PM

[…] Qui potete trovare l’intera descrizione del progetto, lo sketch per Arduino e il programma in esecuzione sul PC, mentre di seguito alcune fotografie durante il suo funzionamento: […]

Monday December 12th, 2011 - 10:03 AM

Hi, thanks for the very useful post. I do have a question: how do I choose the value for the two capacitors?

Monday December 12th, 2011 - 11:31 AM

Hello!
You can use the values (1nF for filtering and 100nF as bypass) suggested in the datasheet.

Wednesday March 21st, 2012 - 11:57 PM

I tried your sketch, and have a .25A load (a lightbulb) and I am not seeing any current whatsoever.

I’m trying to measure current on 120V AC.

Thursday March 22nd, 2012 - 03:28 PM

Hi Patrick,

are you using the same breakout board from SparkFun? Do you see anything on the serial terminal?

Tuesday May 1st, 2012 - 09:53 AM

i have use this sketch for AC power supply, but the value that shown in serial monitor is very fluctuate.

is this sketch for DC only or i can use this sketch for AC?

thanks

Tuesday May 1st, 2012 - 10:27 AM

Hi!

ACS712 is suitable for DC and AC… it gives you only the “actual” measure: as you can see my sketch calculate a mean of 1000 samples… try to increment that number and let me know!

Wednesday June 6th, 2012 - 12:13 PM

I want to know about the low value that we can measure with ACS712. For example I need to measure a mobilep phone current, near to 20mA.
Is it possible?

Thanks

Wednesday June 6th, 2012 - 07:00 PM

Hi!

Of course you can… just add an OpAmp to increase ACS712 sensitivity, or buy this breakout from SparkFun:
http://www.sparkfun.com/products/8883

bye

Wednesday June 27th, 2012 - 01:25 PM

Hello,

I think that this sketch can’t be used to sense AC. The ACS712 will output the measure, but as the incomming current is bipolar (sometimes +I and sometimes -I), the value will change up and down. If you make an average, the result will be always around 0A (or 2.5V). I’m lookiing for another solution… 🙂

Saturday June 30th, 2012 - 06:33 PM

hi luca, I’m from Brazil. Thansks for your help. The project works well!

I have just one more question, please!

Why we use (-13, 51)?

Thanks again I’m very happy!

Sunday July 1st, 2012 - 12:20 PM

Hi Vinicius!

On ACS712 datasheet you can find the “sensitivity” value, which is 185mV/A. Sensor’s output for 0A (= no current) is 2.5V, so if you need the current value, you have to substract 2.5/0,185 which is 13.51.
bye

Thursday July 5th, 2012 - 09:13 PM

Luca Hello!

Please check this software, maybe it’s good for you.

Very easy to use and do not reset the Arduino every time you turn on or turn off the serial port.

Bye

Thursday July 12th, 2012 - 02:03 AM

Just to clarify, you’re working with a DC load not an AC load?

Doesn’t seem like this sketch would work for AC.

Friday July 13th, 2012 - 07:22 PM

Yes: the sensor works also with AC load but you have to change the sketch…

Monday September 17th, 2012 - 07:59 PM

Hi Luca,
This all makes sense except I can’t understand why it will give values below -5A and above +5A:

If you put the value 0 in your equation for units, you get -13.51 out. If the sensor can only sense between -5 and +5, how do we get all these values outside that range?

Thanks

Saturday September 22nd, 2012 - 02:34 PM

Hi Steve,

Have a look to ACS712’s datasheet, page 7: you can find the chart “output voltage vs sensed current”. The output voltage never goes below 1.5V so Arduino’s read value will never be 0. That means you’re not using the complete range of Arduino’s ADC and that’s the reason why SparkFun added an opAmp.

bye

Monday September 17th, 2012 - 11:12 PM

Hey there,

Great tutorial. I have one question though. I am using the breakout from sparkfun:

https://www.sparkfun.com/products/8883

and I am having a problem working with the gain and vref pots.

I dont quite have a good understanding of this. Could you please explain this to me.

This is for my design project at my universtity which is due in a few days time.

sheldonreddy@gmail.com

Saturday September 22nd, 2012 - 02:41 PM

Hi Sheldon,

SparkFun added an opAmp to their board to let you choose the output range of ACS712.
They suggest you this calibration method:
– connect the board to Arduino and read output value without any current flowing in the sensor… move Vref pot to choose the “offset” value (for example move it until you read 512)
– now use a known current value, let’s say you need to measure a max of 1A: use this value and move gain pot until you read 1023 (which is the maximum value Arduino’s ADC can read)
Now your sensor can read -1A – 1A with the maximum available resolution (1024 steps), each step is 1/512 = 0,002A

bye

Monday October 15th, 2012 - 08:55 PM

Ciao Luca
Ho realizzato il progetto ma mi sono reso conto che la risoluzione di lettura della corrente non è sufficiente, quindi vorrei acquistare la breakout board con operazionale incluso per rilevare valori dell’ordine di pochi mA. Dal sito ufficiale leggo che per diminuire il rumore dovuto all’alto guadagno la larghezza di banda viene ridotta da 80 KHz a 34 Hz. Questo influisce sulla frequenza di acquisizione del dato? attualmente utilizzo 1000 rilevazioni al secondo mediate, come da tuo sketch, e non vorrei diminuire. Ti ringrazio per l’attenzione e per quanto ho trovato utile il tuo lavoro. Federico

Tuesday October 16th, 2012 - 04:26 PM

Ciao Federico,

sì, influisce: 34Hz significa che il chip non sarà in grado di “leggere” variazioni nell’intensità della corrente più rapide di 1/34 di secondo. Questo significa che campionare a 1KHz (= 1000 campioni al secondo) è inutile, puoi abbassarlo appunto a 30 circa, che comunque ritengo sufficienti visto che eseguirai una media tra tali campioni.
Fammi sapere come va!

Wednesday October 17th, 2012 - 02:27 PM

Ciao,
Luca complimenti per il blog ricco di spunti interessanti. Ho da poco acquistato l’ACS714 perché avrei in mente di monitorare la corrente assorbita in casa. La domanda è la seguente il criterio per determinare il valore corretto di corrente è lo stesso anche a 230 Volt in corrente alternata?

Thursday October 18th, 2012 - 09:06 AM

Grazie Pixel! Sì, il sensore è pensato anche per corrente alternata e dal datasheet leggo che supporta tensioni > 230V.
Fammi sapere come va!

Thursday October 18th, 2012 - 04:06 PM

Ciao,
grazie per la risposta… ho un dubbio relativo alla mirura del valore medio. In giro ho letto che la lettura della corrente in alternata deve essere effettuata facendo la media un pò diversamente. Tu hai qualche informazione aggiuntiva da darmi? Grazie

Friday October 19th, 2012 - 07:48 AM

Ciao! In effetti per la corrente alternata normalmente si utilizza il valore RMS, che devi calcolare prendendo il valore di picco e dividendolo per la radice di 2… preparerò un esempio di sketch!

Friday October 19th, 2012 - 08:08 PM

Mitico….
terrò d’occhio il blog…

Grazie e Saluti

Saturday October 27th, 2012 - 01:45 PM

Hi when I use the sensor i get:

2.35v = 0A
2.36 = 0.06A
2.37 = 0.1A 511 512 514 517
2.39v = 0.2A 515 522
2.41v = 0.3A
2.42v = 0.4A
2.44v = 0.5A
2.46v = 0.6A
2.48v = 0.7A

The values in the 500’s being the raw analog pin readings that seem to jump around. This sensor is the 5A version and i thought it was supposed to be 2.5v to 5v ie 512 to 1023 0 – 5A. At present it seems really restricted and notchy due to its wavering inaccuracy. Is this broken? Kind Thanks

Sunday October 28th, 2012 - 11:52 AM

Hi!

your sensor has a sensitivity of 185mV/A, so at its maximum (5A) you should read an output voltage of 3,425V: take a look at its datasheet, you can find a chart that plots A/Vout. With 0A you should read 2.5V, your value (2.35) seems to low; did you use the correct capacitors?

Sunday October 28th, 2012 - 05:10 AM

ciao luca, i’m using acs712 30A with your sketch (11.54Vdc and 12Vdc 10W globe as a test) without editing your sketch I get a reading of 0.30 which seems to low, i’m calculating 0.83, but when i edit the sketch to change sensitivity to 66mV/A i get 10.03A
where am i going wrong?

Monday December 3rd, 2012 - 04:02 PM

Salve luca,

sarebbe possibile utilizzare secondo te l’ACS 712 per rilevare la corrente in uscita al differenziale di casa tramite arduino? O è piu consigliato un dispositivo meno invasivo, quale ad esempio una pinza amperometrica, sempre pilotata con arduino? Ovviamente preferirei la scelta piu economica…e piu efficiente, dato che vorrei usare il sistema per monitorare l’uso di corrente in casa ed evitare che scatti il contatore, magari sfruttando un avviso acustico (sempre pilotato da arduino).

Grazie mille in anticipo

Thursday December 6th, 2012 - 03:39 PM

Ciao Gianfranco,

io sarei, viste anche le tensioni e correnti in gioco, per un sensore ad effetto Hall, tipo:

http://www.seeedstudio.com/depot/-p-519.html?cPath=144_154

fammi sapere come va!

Wednesday February 20th, 2013 - 03:42 PM

What is the range of values displayed in the serial monitor of arduino?

Wednesday February 20th, 2013 - 04:02 PM

Hi Paul,

it depends on the sensor you’re using… ACS712 is rated -5/+5A

Wednesday February 20th, 2013 - 05:18 PM

My serial port value is not varying with the load current.It is remaining a constant value.I think it is not the problem of my circuit.I had checked the circuit several times.Can you suggest a solution for this ?

Wednesday February 20th, 2013 - 05:20 PM

Paul: try reading the “raw” output value of the sensor with a multimeter: if it does change it could be a problem of the analog pin of your Arduino

Wednesday February 20th, 2013 - 05:47 PM

i had tried that.But the multimter value is not changing.It remains constant.What may be the problem?

Thursday February 21st, 2013 - 04:10 PM

post a schematics of your sensor… how are you changing the current in it? which value do you read on your multimeter?

Wednesday February 27th, 2013 - 10:04 AM

excellent work….keep it up friend…thankyou…:)

Saturday March 9th, 2013 - 02:47 AM

i have a 5a model of acs712 sensor. i had tried use the equation(above) to find current value. if the adc value is 1024, result is 13.49a. but result should be 5a for 1024. where do i make mistake can you suggest anything about this situation?

Saturday March 9th, 2013 - 10:12 AM

Hello! With 5A, the ACS712 sensor won’t output 5V (look at its datasheet, pag.7), but a value near 3.5V that analogRead() reads 711.

Tuesday April 2nd, 2013 - 06:36 PM

Hello luca,
Thank you for this very helpful post.
I’d like to measure AC current, but I can not find a sketch in internert and do not get it myself.
Can you or anyone else please give me such a sketch.

Monday April 8th, 2013 - 08:01 PM

Hi Nikita,

the sensor measures the “actual” current, so for AC you need to decide which “value” do you want to measure.
You usually want the RMS value, that – assuming your wave is a true sine – is 0.707 times the peak value… so you need to write a sketch that measures several values, find the maximum (= peak) and calculates the RMS value.

Sunday April 7th, 2013 - 01:34 AM

Hello, very good design.

I did the same thing in my project that only the current sensor captures a peak current and then back to 0, I do not know what’s going on, I think he gets the current instant. Do you think my sensor may be the problem? he is not the SparkFun.

Monday April 8th, 2013 - 08:06 PM

Hi Andre, is the current in your project alternate?

Sunday April 21st, 2013 - 07:52 PM

Hello Luca,

I am want to measure the current of a fan, I used the same example but its current returns is zero, to measure AC is different?

Friday April 26th, 2013 - 02:07 PM

Hi Andre,

AC current rapidly changes, so you have to decide “which” value you want to measure. Usually, they choose the RMS value, which is the maximum value / sqrt(2). I’m going to write a tutorial about it, so stay tuned!

Tuesday May 7th, 2013 - 02:55 PM

Ciao,
ti scrivo perchè c’è qualcosa che non mi torna molto.

Tu scrivi questa formula:
A = 0,026 * units – 13,51

Il sensore riesce a leggere tra -5A e 5A.
units varia tra 0 e 1023.
quindi in linea di principio quando esce 1023 dovrei avere 5A.

Per cui faccio questa prova sostituendo a units il valore 1023 nella tua formula è il risultato ottenuto è: 13,088A differente da quei 5A che mi aspettavo.

C’è qualcosa che non torna oppure è tutto giusto?

Se è tutto giusto avendo il sensore da 20A questa formula è corretta?
A= (0.0049 * units – 2,5)/0,1

Tuesday May 7th, 2013 - 06:45 PM

Ciao Pier

l’errore nel tuo assunto è che l’uscita del sensore, per 5A, sia 1023 ovvero 5V. In realtà se guardi il grafico sul datasheet, vedrai che la sensibilità del sensore è 185mV/A, centrata a 2.5V. Quindi con 5A di corrente avrai una tensione in uscita di 3,425 e una lettura dell’ADC di Arduino di 712.

Wednesday May 8th, 2013 - 02:15 PM

e quindi per il sensore da 20A sapendo che la sensibilità è 100mV/A è giusta?

A= (0.0049 * units – 2,5)/0,1

Monday May 13th, 2013 - 09:49 AM

yes!

Saturday May 11th, 2013 - 04:19 AM

Hello Luca,

I bind AC load sensor and the connecting pin to pin Vout analog Arduino, the value read at the input A0 would be the current through the sensor. Am I right or wrong? When I put a load sensor my AC does not alter the value of output (it works as if current = 0).

For load DC the sensor works correctly.

Monday May 13th, 2013 - 09:39 AM

Hi Andre

with AC loads, the output of the sensor should rapidly change following the voltage. Which value do you get from your A0 pin? Try reading it many times…

Monday May 13th, 2013 - 06:00 PM

Ciao Luca
ho seguito il tuo consiglio e acquistato la breakout board di sparkfun con stadio amplificatore, però non riesco ad effettuare una misurazione stabile.
Il mio obiettivo è quello di misurare affidabilmente valori di corrente nell’ordine dei pochi mA, con una risoluzione massima di 1mA, meglio se 0.5mA. Credi che sia un traguardo raggiungibile? Ho seguito le istruzioni per la calibrazione del sensore utilizzando i tuoi valori di riferimento (0A e Vref a circa 512bit – 1A 1023bit). I problemi sorti sono due:
– Una volta terminata la calibrazione il valore calcolato con la tua formula, senza alcuna corrente nel sensore risultava di circa mezzo Ampere. Utilizzando il sensore amplificato devo utilizzare una formula diversa da quella espressa sopra?
– La lettura dei valori risulta sempre molto instabile, infatti i 512 bit di calibrazione non sono stati molto accurati. Questa fluttuazione è normale?
Ti ringrazio in anticipo, e complimenti per il tuo impegno in questo blog e nell’aiutare la comunità.

Tuesday May 14th, 2013 - 08:44 AM

Ciao Federico,

purtroppo no, non è raggiungibile a causa delle caratteristiche fisiche del sensore (guarda le FAQ del produttore). Utilizzando cmq il modulo amplificato (a cui dedicherò un prossimo articolo) la formula va adattata in base ai settaggi del potenziometro.

Tuesday May 14th, 2013 - 01:52 PM

Grazie Luca
purtroppo per le mie esigenze dovrò optare per uno strumento decisamente più costoso, ma apprezzo molto la tua dedizione.
Continuerò a seguire il tuo blog, in attesa del prossimo progetto 😉

Saturday May 18th, 2013 - 12:40 PM

if my load is a resistor and i connect this sensor in parallel in as much its small internal resistance most of the current of the source will pass in this sensor and it will be maximum and that mean it won’t be the real load current when i remove the sensor

Sunday May 19th, 2013 - 08:07 PM

Hi!

Thursday May 30th, 2013 - 10:09 PM

Ciao Luca,
ti scrivo perché ho comprato quattro di questi sensori: 1 da 5A f.s, 1 da 20A f.s, 1 da 30A f.s, e 1 da 50A f.s., con la convinzione che bastasse collegarli ad un voltmetro da qualche Volts fondo scala per poter leggere una corrente senza bisogno di utilizzare un amperometro dotato shunt. Ho costruito degli alimentatori e dei caricabatteria, e volevo misurare la corrente erogata con un amperometro che non avesse shunt. Di qui la scelta dei sensori. Ebbene mi sono sbagliato in pieno, dopo aver provato i sensori ho dovuto constatare che questi circuiti si limitano generare una tensione su un piedino d’uscita che decresce con l’aumentare della corrente che scorre al suo interno e, con un voltmetro, si misura soltanto una tensione variabile ma che non ha nulla a che vedere con un valore di corrente.
Ora, ti chiederei la cortesia di fornirmi il tuo parere per sapere se è possibile interfacciarli con dell’elettronica ed utilizzarli abbinati ad uno strumentino digitale da pannello, oppure devo buttarli in un cassetto.
Naturalmente non mi riferisco ad Arduino, perché non so neppure cosa sia.
Spero di poter leggere una tua gradita risposta.
Grazie e cordialità.

Saturday June 8th, 2013 - 12:32 PM

Ciao Massimo

i sensori restituiscono in uscita una tensione che è “linearmente proporzionale” alla corrente che li attraversa (guarda il grafico sul datasheet). Per poter calcolare la corrente una volta letta la tensione devi però effettuare qualche “calcolo”, usando la formula A = (V – 2,5) / 0.185, quindi ad esempio se leggi 3V, la corrente che attraversa il sensore sarà circa 2.7A.

Friday June 14th, 2013 - 02:51 PM

Ciao Luca,
ho letto questo tuo scritto svariato tempo fa quando ho cominciato a pensare al mio progettino per monitorare tra l’altro i consumi elettrici di casa. Già allora lessi che la tua formula andava bene per la corrente continua ma che bisognava fare altri calcoli per la AC. Da quel che ho letto nei vari reply non è ancora esplicitato come dovrebbe essere il calcolo secondo te quindi, col tuo permesso, incollo qui il metodo che ho sviluppato io e che è attualmente operativo sul mio arduino con risultati accettabili.

*********************************************
float watt0; // Potenza letta dal sensore su A0

void loop(){

watt0 = rilevaConsumo(sensorPin0); // Rilevazione dal Sensore A0

}

// Lettura dei dati dei Sensori ACS712 20A
// Le letture istantanee del sensore da parte di Arduino forniscono quello che ho chiamato il numero di Step.
// Dallo Step posso ricavare Volt e Ampère (basandomi sui dati del Datasheet) ma dovendo leggere un valore Alternato (corrente Alternata)
// è inutile che faccio calcoli per determinare una grandezza istantanea che non mi serve.
// Per calcolare il Consumo ( P = V * I * cosFi) ho bisogno di ricavare dalle letture il Valore Efficace della Corrente.
// Anziché calcolare la corrente istantanea ad ogni lettura e poi estrarne il valore efficace ho calcolato il valore efficace delle letture
// e con quello ho poi calcolato il Valore Efficace della Corrente.
// Ovviamente il calcolo prevede una certa approssimazione poiché la tensione non è quasi mai esattamente 220V e dopo
// un po di test ho adottato un cosFi di 0,9

float rilevaConsumo(const uint8_t sensore){
// float lettura;
float totLetture;
// double rmsLetture;
// float ampere;
for(int i = 0; i < 1000; i++) { // 1000 letture sono un campione più che sufficiente di letture.
// lettura = analogRead(sensore) – 511;
// rmsLetture = rmsLetture + sq(lettura); // Sommatoria del quadrato delle letture.
totLetture = totLetture + sq(((float)analogRead(sensore)) – 511); // Sommatoria del quadrato delle letture.
delay(1);
}
// float risultato = 220 * ((float)((double)sqrt(totLetture/1000)) * .048828) * .9;
// Serial.println(risultato);
// rmsLetture = sqrt(totLetture/1000); // Valore Efficace delle Letture (Radice Quadrata della Media del Quadrato delle Letture)
// ampere = rmsLetture * .048828;
// return 220 * ampere * .9;
return 220 * ((float)((double)sqrt(totLetture/1000)) * .048828) * .9;
}
*********************************************

Monday June 17th, 2013 - 09:16 PM

Ciao Alessio!

direi che è ottimo: grazie per averlo condiviso!

Friday June 21st, 2013 - 06:07 AM

i already desing a sketch for ac , it works, im working to make it better its some one need help, just ask me at araujo.inl@gmail.com

Tuesday September 10th, 2013 - 05:55 AM

HI
I AM INTERESTED ARDUINO.
I TRY ACS712 CONNECT TO 220 VOLTS FOR MEASURE CURRENT WATER PUMP 1 PHASE ,1 HP, BUT I AM SELF CONTAINED . I LIKE KNOW CALCULATE FORMULA CURRENT FOR ARDUINO FOR ACS712 FOR AC CURRENT .
TANK YOU

Tuesday September 10th, 2013 - 08:13 AM

Hi Mehdi,

for AC current, you usually want the “RMS” value, which is the peak value / sqrt(2)

Thursday September 12th, 2013 - 03:03 PM

very nice design.
but i would like know what is the use of the samples

Tuesday September 17th, 2013 - 09:08 AM

Hi Israel, you need to grab many samples if the waveform is variable (like the AC) to find the max or average value

Friday September 20th, 2013 - 09:21 AM

Hi Luca,

Thank you for a very good page. I have a ACS712-20A (sensitivity is 100mV/A) connected to a PIC16F818. Al circuit connections are correct. A powersupply is connected to the ACS712 through a 10R 10W resistor. I always get a reading which is about .06 too low. Formula used is (reading – 511) * 49. Any help please ?
Thank you.

Tuesday September 24th, 2013 - 07:22 AM

Hi, luca,
thank you for your page. i think sensor need current more than 1 amp.
i think circuit should be current more than 1 A.
FOR EXAMPLE : ADAPTER OUTPUT 2 A ,9 V —
IS IT CORRECT ?

Tuesday September 24th, 2013 - 12:40 PM

Hi Mehdi,

the sensor I used can measure up to 5A but the ACS family has also ICs for 10-20A

Tuesday October 8th, 2013 - 12:46 PM

Hi luca
Just for making sure , will this ic work at 220V AC .

Tuesday October 8th, 2013 - 01:00 PM

Yes, it works fine!

Saturday October 12th, 2013 - 08:02 AM

HI
I AM TRY FOR ACS712 GET CURRENT BUT I CAN NOT IT .
FOR EXAMPLE :LIGHTING DC -12 VOLTS , I =3.2 A MESSURE MOLTIMETR.
BUT WHEN CONNECT TO ARDUINO SERIAL ——–>NUMBER RESULT 0.02 mA .
I CONFIUSE .

Tuesday October 22nd, 2013 - 06:10 PM

@MEHDI Wrong formula i would guess… Maybe you should write less caps but more content – this would make helpful guesses more likely 😉

Tuesday December 3rd, 2013 - 08:28 PM

Sir the same code can work for acs714?

Wednesday December 4th, 2013 - 08:51 AM

Hi! Yes, it has the same sensitivity (185mV).

Saturday December 7th, 2013 - 11:50 AM

Ciao Luca, qualche post fa Alessio aveva postato lo sketch per la misura della potenza istantanea approssimata.
Potresti ripostarla in maniera più semplificata e completa?
Grazie

Tuesday December 31st, 2013 - 02:20 PM

Ciao Davide,
quale è la parte che ti risulta più ostica?
Di quello che ho incollato la maggior parte è commentata. In realtà solo poche righe (quelle non commentate) sono il risultato finale.
Se posso esserti d’aiuto, son qui.

Alessio ;0)

Monday January 13th, 2014 - 07:00 AM

Can you give me another great tutorial on how measuing AC current with the acs712? Thanks

Thursday January 30th, 2014 - 11:56 PM

Ciao Alessio, ho provato il tuo sketch e mi restituisce una serie di errori in fase di compilazione.
L’ho riscritto in questo modo, soltanto per riverale la corrente, viene compilato ma il risultato non è soddisfaciente, è una specie di contatore in avanti.
Inoltre potresti spiegarmi il significato del comando ‘sq’ ?
Io ho scritto questo:
#include
LiquidCrystal lcd(12, 11, 53, 51, 49, 47);

float Sensore = 0;
float totLetture;
float Corrente = 0;

void setup() { ; }

void loop() {

for(int i = 0; i < 1000; i++)
{
totLetture = totLetture + sq(Sensore);
Corrente = (sqrt(totLetture/1000)) * 0.07398;
delay(1);
}

lcd.begin(16, 2);
lcd.setCursor(0, 0);
lcd.print(Corrente);

}

Thursday February 13th, 2014 - 10:23 AM

Ciao Davide,
sugli errori di compilazione non so cosa dirti, di tutto quello che ho postato sopra in realtà le righe utili non commentate dovrebbero essere solo queste:

float rilevaConsumo(const uint8_t sensore){
float totLetture;
for(int i = 0; i < 1000; i++) { // 1000 letture sono un campione più che sufficiente di letture.
totLetture = totLetture + sq(((float)analogRead(sensore)) – 511); // Sommatoria del quadrato delle letture.
delay(1);
}
return 220 * ((float)((double)sqrt(totLetture/1000)) * .048828) * .9;
}

se chiami questo metodo passandogli l'identificativo del pin a cui è collegato il sensore dovrebbe restituirti la potenza assorbita in quel momento.

Il Valore Efficace di una grandezza sinusoidale è dato dalla Radice Quadrata della Media del Quadrato delle Letture.
Il comando sq() eleva al quadrato il valore messo tra parentesi. Quindi con questo comando eseguo il quadrato delle letture che poi sommo nel totLetture e poi ne estraggo la media facendo totLetture/1000.
Quindi alla fine del for(), dopo che ho totalizzato 1000 letture faccio il conto finale ovvero la radice quadrata ( sqrt() ) della media del quadrato delle letture ( totLetture/1000 )
Il comando sqrt() fa la radice quadrata del valore messo tra parentesi. Quindi sqrt(totLetture/1000) mi da praticamente il Valore Efficace delle letture eseguite che moltiplicato per .048828 (per il sensore da 20A) o .07398 nel tuo caso, si ottiene il Valore Efficace della Corrente.

In pratica se prendi il mio metodo e cambi il return con:
return ((double)sqrt(totLetture/1000)) * .048828;
dovrebbe restituirti il Valore Efficace della Corrente.

Per quel che riguarda il tuo sketch riscritto, non l'ho provato ma ad occhio direi che il problema del "contatore in avanti" è dovuto al fatto che non azzeri il totLetture prima di entrare nel for(). Questo comporta che totLetture è in continuo incremento (fino al limite del contenibile da un float, dopodiché dovrebbe ripartire da zero).
Comunque, basta che prima del for() metti un totLetture = 0;
Inoltre il calcolo della Corrente (Corrente = (sqrt(totLetture/1000)) * 0.07398;) è inutile all'interno del for(). Mettilo fuori, dopo il for().

Fammi sapere.

Alessio.

Thursday February 13th, 2014 - 10:26 AM

P.S.: Recentemente, a causa in un cortocircuito in casa, m’è esploso uno degli ACS712…. e l’altro ha smesso di funzionare :0/
Passerò a qualcosa di Non Invasivo.

Sunday February 16th, 2014 - 04:11 PM

Ciao Alessio, in effetti con quelle modifiche è tutto ok, non azzeravo TotLetture. grazie.

Sunday April 20th, 2014 - 12:12 AM

I am testing your code as shown, but getting a strange result. For 1A current (with Fluke dvm in series for ref), the Arduino displays 0.733A.
Any guidance appreciated.

Sunday April 27th, 2014 - 12:36 PM

Hi Edward, are you measuring AC or DC?

Wednesday May 28th, 2014 - 02:27 PM

Hey luca,

I have ACS712 +-5A sensor board.

Is it possible to measure 220V AC by connecting multimeter on its pins? I get constant value all time which is 1,46V even when on load.

Wednesday May 28th, 2014 - 08:34 PM

Hi Mehmet

of course if everything is ok you should be able to read a voltage that depends to the load you apply… but the output voltage is AC too, are you reading AC voltages with your multimeter?

Wednesday June 11th, 2014 - 08:04 AM

Hai Luca..
I would like to ask about ACS712. Can this sensor measure current that connected to solar panel with voltage output grater that 20 V DC? My input would be a solar panel and i would like to measure voltage and current it draw using arduino..

Hope to hear from you Luca

Monday June 16th, 2014 - 08:23 AM

Hi, yes it works!

Monday August 11th, 2014 - 01:22 AM

Luca.

I am just about to buy this board for a project I am working on but I am not clear where to hook to the arduino and the instructions I found online were much more complicated then how you have it cleanly hooked up.

It looks like it can easily connect right to the board can you tell me what pins you are hooked to on the Arduino?

Thanks so much and I am sorry for the newb question.

Wednesday August 27th, 2014 - 02:25 AM

Have you made the tutorial for AC current mesurement?

Sunday September 21st, 2014 - 07:24 PM

Luca,
ho eseguito il test seguendo le tue indicazione (caso vuole che ho anche il TRITON vista la passione per il modellismo) e tutto fila liscio.
Lo scopo dell’acquisto del sensore è però quella di tenere sotto controllo il consumo di alcuni elettrodomestici; in particolare ho eseguito un test con un ventilatore tipo scaldo bagno da 2200W max e qui sorgono i problemi; dal sensore rilevo assorbimenti vicini allo 0.0A e questo mi suona strano.
Dove può essere il problema?
ciao grazie in aticipo
Mauro

Monday September 22nd, 2014 - 12:57 PM

Ciao Mauro, il problema potrebbe essere relativo al fatto che stai misurando una corrente alternata quindi a media zero… la corrente in tal caso deve essere RMS (http://it.wikipedia.org/wiki/Corrente_alternata)

Thursday January 22nd, 2015 - 02:14 PM

Voglio misurare la corrente assorbita da un circuito (al momento utilizzo una resistenza 2.1 K come circuito)
Il circuito viene alimentato mediante un trasformatore che fornisce 7.7 Vcc
Il valore in uscita dall’ACS712 viene acquisito dallADS1115 (adc a 16 bit)
il valore restituito a circuito aperto, non alimentato, è 2444.45 mv
alimento il circuito con il trasformatore è ottengo 2410/2411. Dopodiché inverto la polarità è ottengo 2373/2374 mv

Ai capi della resistenza la tensione di alimentazione è di +/- 7.3 Vcc

domande:
1 poiché il valore a riposo non è 2500 mv dovrei calcolare il valore medio a riposo ed utilizzarlo come punto di riferimento ?
2 con l’acs712 posso misurare correnti di segno opposto ?
3 i valori di corrente misurata non dovrebbero essere più o meno uguali ma opposti ?
4 che valori di corrente mi devo aspettare ?

grazie.

Wednesday April 1st, 2015 - 08:31 AM

Ciao Luca, vorrei realizzare un esperimento per dei ragazzi delle scuole medie, in pratica vorrei misurare la potenza in uscita da un pannello solare con Arduino e trasformarla in una versione visiva con una barra LED che si accendono in proporzione alla potenza in uscita dal pannello. Lo scopo è far capire loro che il pannello deve essere orientato correttamente come un girasole…. Riesci a darmi una mano?

Wednesday April 1st, 2015 - 09:28 PM

Ciao Mario! Bel progetto! Il pannello solare esce a 12V CC? Striscia di 10 LED?

Friday April 3rd, 2015 - 10:35 AM

Ciao Luca, grazie della risposta. Il pannello esve a 17V CC alla massima potenza e si barra 10 led.

Friday April 3rd, 2015 - 12:21 PM

Ciao Luca, Complimenti per la tua preparazione.
Vorrei chiederti gentilmente se potessi farmi un pccolo schema con i relativi componenti da usare riguardo questo.
Vorrei avere un ritorno “reale” dello stato della lampadina in un pin arduino, cioè , se la corrente passa la lampadina è integra ed accesa. ovviamente convertendo il circuito in 5v per i pin di Arduino.
te ne sarei grato 🙂
Grazie .

Tuesday April 7th, 2015 - 05:58 PM

Ciao Manu! Se la lampadina è in corrente alternata (ad esempio una comune lampadina a 220V) la soluzione più semplice e non invasiva è usare una “pinza amperometrica” (es venduto da Seeedstudio) attorno al filo e leggere il suo output da Arduino (come spiegato sempre nella wiki di Seeeds).

Friday April 24th, 2015 - 05:48 AM

hy luca,
in my project i used arduino , xbee and acs712_05.
i will monitoring current in arduino and xbee. the specifications and arduino xbee consumes a current of 40 mA. so that current flows at – + 80 mA. but I use a current sensor with programming that you can provide 150 mA. whether there are any of my measurements ??

Friday April 24th, 2015 - 09:07 AM

Hi! Unfortunately the sensitivity of the sensor itself is not enough to measure a such small current… you need a different module (give a look to the one that Sparkfun sells, it has an OpAmp to increase the sensitivity)

Friday April 24th, 2015 - 09:59 AM

but when I measure current 40mA same audio only using avometer. sedangkat when using xbee results are much different from the current sensor reads 150mA while using avometer of 75mA.

Tuesday May 19th, 2015 - 01:36 PM

Luca,
does your solution work on arduino due or just on uno?

Wednesday May 20th, 2015 - 07:57 AM

Hi! It should work but I don’t own a Due to test the sketch…

Wednesday July 8th, 2015 - 10:04 AM

Ciao Luca.

Devo realizzare un piccolo sistema di condizionamento per un segnale di corrente. Si tratta di una corrente che alimenta un motore AC a 220V (una pompa) il cui assorbimento nominale è di circa 250mA. Devo generare un paio di segnali (TTL) che indichi se la pompa è accesa o sotto sforzo. Ho stimato una corrente di sovraccarico di 500mA. Avevo pensato ad una ACS712, ma viste le correnti in gioco non credo sia idoneo (rischio di usarlo al limite della sua sensibilità!).

Ti ringrazio!
Federico

Thursday July 9th, 2015 - 05:11 PM

Ciao Federico, puoi usare il MAX712 con un opamp esterno come nella breakout board di Sparkfun oppure altri sensori pensati proprio per low currents, esempio questo.

Tuesday August 11th, 2015 - 01:13 PM

hi luca
Do you test your approach by DSP(F2812)?
How i test your approach in my project (measurment current PMSM motor) with DSP(F2812)?

Thank you.

Wednesday August 12th, 2015 - 01:16 PM

Hi! No, I simply used an analog pin and did all the maths in Arduino… sorry but I can’t help with TI DSPs

Saturday October 3rd, 2015 - 01:29 PM

Ciao Luca, complimenti per gli articoli molto chiari, mi sono letto tutto il sito 😀
Avrei una domanda, ho questo sensore ma nella versione a 30A, se il zero point è 2,5v e per ogni Ampere ho 0,66mv per 5V dovrei avere 30A di carico ma calcolando la formula mi risultano solo 3,7A. Potresti aiutarmi?

Tuesday October 6th, 2015 - 08:01 AM

Ciao Francesco, questi sensori non raggiungono il “fondo scala” (5V) per il valore massimo di corrente ma appunto hanno un incremento di 66mV per ogni A, quindi al massimo avrai 2.5V (zero point) + 30A * 66mV/A = 4,48V.

Friday January 29th, 2016 - 09:49 AM

hi luca ,
i am using acs712 20A current sensor,when i measure the output voltage is 2.30 using multimeter.
i am not getting same value in serial monitor.this is my code

[…]

ouput from serial monitor is

sensor value is 730
voltage is 3.58
current is 5.773

Sunday January 31st, 2016 - 01:29 PM

Hi! Are you measuring with your multimeter the voltage drop between the analog pin and ground? It’s very strange a such different value (2.3V vs 3.58V), the analog pin is usually very accurate! Try measuring the current with your multimeter, which value do you read?

Saturday April 9th, 2016 - 08:27 AM

Grazie, mi è stato molto utile! 🙂

Monday May 16th, 2016 - 08:11 PM

Sir i want to use controller instead of arduino to measure A.C current, what changing i have required to get correct answer.Kindly inform me.

Tuesday May 17th, 2016 - 11:19 AM

Hi! The sensor measures the instantaneous current… if AC, you need a “way” to calculate it… I usually log some measurements and compute their RMS value (https://en.wikipedia.org/wiki/Root_mean_square)

Thursday July 7th, 2016 - 11:43 PM

Luca – thanks for a nice chat on the acs712. I would like to know if you have done the write up on using the chip on AC current. I am battling with a part of my project to protect a AC motor from over current.

Monday July 25th, 2016 - 09:22 AM

Hi! The sensor works fine with AC, you just need to decide “which” (RMS, max…) value of current you’re interested on.

Saturday April 8th, 2017 - 09:59 PM

Hi Mr. Luca,

i want to measure AC load upto 20A and I’m not getting design for this AC..

Monday April 10th, 2017 - 08:53 AM

Hi Dinesh: the sensor returns the “actual” current value… for AC you have to decide “which” value you want (RMS, max…) and do some maths on the samples you get from the sensor.

Sunday April 23rd, 2017 - 04:08 PM

Buongiorno Luca, è possibile usare ACS712 per misurare la corrente erogata da una batteria LiPo? Ho connesso l’uscita della batteria direttamente ai pin di ingresso di ACS e poi ho connesso il sensore a 5V-A0-GND come da tua spiegazione. E’ corretto procedere in questo modo?
Grazie e complimenti per il sito.

Friday June 30th, 2017 - 12:53 PM

Ciao, secondo me c’è un errore… possibile che il “/1000” vada fuori dal ciclo for? se no così divide per 1000 ad ogni incremento di i

Friday June 30th, 2017 - 03:36 PM

ciao, per evitare di tenermi tutti i valori, calcolo una moving average ad ogni campionamento

Sunday January 28th, 2018 - 09:04 AM

Ciao, avrei bisogno di realizzare qualcosa che accenda un Led se c’è passaggio di corrente (12v). Posso utilizzare questo sensore? Posso evitare do usare arduino e sfruttare solo questo? Quale versione devo usare? Grazie

Monday January 29th, 2018 - 10:22 AM

ciao Daniele, il sensore dipende da quanta corrente passa… questo sensore non è in grado di pilotare direttamente un led, devi cmq aggiungerci qualche “logica” (arduino…)

Wednesday February 21st, 2018 - 09:42 AM

Ciao Luca,
ottimo articolo complimenti. Una curiosità:leggevo che questi sensori Hall a riposo misurano 2,5v, poi il valore cresce o decresce a seconda del verso della corrente. Vale lo stesso anche per la corrente AC?

Monday February 26th, 2018 - 02:38 PM

Ciao Sandro, sì questi sensori funzionano egualmente con la corrente AC